Walk in Battery Altitude Test Chamber
Walk in battery altitude test chamber is used to simulate test environments at different altitude, nd can be customized according to test requirements. This is especially important for defense and aerospace applications where pressure and vacuum conditions are worse,such as aviation, aerospace, communications, electronics and electrical applicances. The test objective is to ensure the maximum possible quality and reliablity of the product or material at different heights.
ASTM D6653 TEST PROCEDURE
||1 ~ 100Kpa
||Touch Screen Controller
|Vacuum unit switch
||RT ~ 0℃
||Rise rate 305m per 30-60s until setting pressure is achieved.
|Pressure holding time
||SUS# 304 Stainless Steel
||Steel plate with paint coating
||Single phase, 220V/50HZ
||ASTM D4169 Schedule I, ASTM D6653
Step 1: Select your vacuum chamber which capable of appropriately withstanding 1 atm pressure differential across the walls. The vacuum gauge must be sealed to the chamber and capable of a vacuum range from 0 to 100 kPa.
Step 2: Determine your sampling method which should be representative of your sample to permit adequate determination of performance. If you cannot determine a good sample size, at least 3 representative samples must be selected for performance evaluation.
Step 3: Test specimen must be conditioned to 5.6 +/- 2 Degrees Celsius for 24 hours prior to testing. Test specimens must also be tested at 5.6 +/- 2 Degrees Celsius.
Step 4: Place the Test Specimen into the Vacuum Chamber and turn the vacuum source by making the vacuum rise by 1000 ft or 305 m for every 30 to 60 seconds until the appropriate pressure/vacuum is achieved.
Step 5: Attain a pressure equivalent of 4,877 m +/- 5%; this correspond to an absolute pressure of 56.3 kPa, or 16.22 in Hg. An elevation of 4,267m may be aimed for if 4,877 m is not achievable or inappropriate; this corresponds to an absolute pressure of 59.5 kPa.
Step 6: Maintain the achieved vacuum for 60 minutes.
Step 7: After the 60 min vacuum hold time has elapsed, release the vacuum at a rate of 1000 ft or 305 m for every 30 ~ 60 seconds.
Step 8: Once vacuum has been released, open the chamber, and examine your specimen for damage or deformation.
SEE WHAT COMPANY YOU WILL WORK WITH?
1.Professional Manufacturer of various kinds of testing equipment over 15 years, defect rate is controlled under 0.02%.
2. Completely Production Line. From Raw Material to Delivery, all links are under our control. The quality can be assured.
3. ISO 9001:2008 Quality Management System Certificate.
4. OEM/ODM Service. Just tell us your idea! Our R&D Team is standing by…
5. Our Marketing Investigation Department will send Marketing Research Report to you each season.
6. We are trusted by SCS, INTERTEK, HST, COFCO, ESC, KBMC, LENOVO, PANASONIC, EMC, etc.
7. We accept 100% inspection.
8. Guarantee Period: 365 days, After Sales Service Department provides professional service with you, any questions, just contact us.