Is The Humidification Of The Constant Temperature And Humidity Test Chamber Abnormal?
Humidity is an indispensable part of the constant temperature and humidity testing machine. Many users will encounter the sudden failure of normal humidification when operating the test chamber on a daily basis. This situation is not necessarily caused by equipment quality problems, but may be caused by incorrect operation. Come down, you may try the following methods to check the reason why the constant temperature and humidity testing machine cannot humidify normally, and help solve the problem in time.
1. Check whether the water tank is short of water. If there is a lack of water in the water tank, the test chamber will not be able to humidify properly. The treatment method is to directly pour water into the water tank. If you do not know where the water is added to the equipment, you can consult the manufacturer of the test chamber you purchased. But the premise is that it must be explained with the manufacturer whether it is a standard chamber or a small test chamber, because the location of adding water is different for the two. When it is a device that automatically adds water, this possibility can be skipped directly.
2. Check whether the warm ball gauze of the constant temperature and humidity chamber is normal. If the wet ball gauze of the constant temperature and humidity test chamber lacks water or is very dirty, in fact, the device should be able to humidify normally in this case, but it cannot be displayed on the control instrument. When encountering such a problem, you only need to change a new gauze or clean it. In addition, when reinstalling, you should pay attention not to attach the gauze to the temperature sensor of the device, so that the humidity displayed by the meter is still not correct. precise.
3. Check whether the water tank of the constant temperature and humidity testing machine leaks. In fact, water leakage from the water tank is also one of the reasons why the constant temperature and humidity test chamber cannot be humidified normally, but this problem generally rarely occurs. But if it is really caused by this problem, you must contact the manufacturer in time for after-sales processing.